toggle quoted messageShow quoted text
Thanks for confirming the operation of the OCXO.
How critical is 5 volt precision in using the OCXO?
On Mon, 24 Sep 2018 10:16:38 -0400, Graham <firstname.lastname@example.org> wrote:
Your brief description of the operation of your OCXO sounds like a
typical OCXO - higher current at start up til stabilization, then lower
current draw. Same goes for the frequency drift.
The adjustment only needs a potentiometer to be used as a voltage divider
and is not overly critical as far as value is concerned. I tend to use
something around 5k.
There is often a Vref output from OCXO that you would use on one side of
the pot rather than the VCC, the other side of the pot to ground, and
the wiper to the ADJ.
Don't expect a lot of adjustment range, on the order of single digit Hz
+/- of the nominal frequency
cheers, Graham ve3gtc
On 9/24/2018, "Alan Jones" <email@example.com> wrote:
I finally received a OCXO from the extremely slow boat from China.
How much resistance do I need in order to adjust "Vcont" on pin #3 ?
My initial hook up without Vcont, according to my eBay 8 digit frequency
counter showed a slow frequency drift from about 9.999.820 MHz up to
10.000.027 and stabilized after about 7 minutes. Current draw got up to a
max of around 400 mA and then settled in around 198 mA and the oscillator
can was warm to the touch.
The OCXO I received from eBay was used and has a calibration sticker on it.
While looking for info on this OCXO I stumbled onto this document about
Precision Oscillator Overview.
I just ordered the GPSDO that Chuck referenced in an earlier email.
We will see how long it takes to get here!