Re: Insertion delay needed for BNC-BNC couplers UG491A/U (M-M) and UG914/U (F-F)
Hi Johntoggle quoted messageShow quoted text
You are doing some calculations which is not reflecting the physics. The BNC adaptors characteristics dependent on the mating part. When both are fitted with isolation (PTFE) then the velocity factor is 0.695. Bur the SDR Kits male short adaptor is fitted with isolation but the open is not. You are trying to involve the female short into the calculation but that is so the delay thru the FF adaptor terminated with the male short is not just the summed delays. E.G. the calculated value 77.11ps is not the same value as the S21 thru delay which is 72.45ps. It is far more complex and these delays you use are simple delays which is the average delay over the frequency span. In the arbitrary settings you will see that there is a long expression which is the frequency dependent delays for the male and female short and open. Load the female kit and create a custom trace with the expression s_open as well as s_short and right click on the custom trace, select copy and paste the copy to S11. Then used extension port delay and trim delay to a horizontal phase line and you will se it is not possible as it is very frequency dependent. So you cannot claim a VF=0.67 due to this, sorry to say it is not that simple to find the male short delay based on mechanical dimensions. The internal structure of the female female adaptor is also not ideal as there sare sections not filled with PTFE that is why I have so implemented so complicated expressions in the cal kit definitions.
Fra: VNWA@groups.io <VNWA@groups.io> På vegne af John KN5L
Sendt: 23. maj 2020 01:23
Emne: Re: [VNWA] Insertion delay needed for BNC-BNC couplers UG491A/U (M-M) and UG914/U (F-F)
Yes, 3.333... is correct value. Normally I use VNWA port extension window to compute physical length from time.
SDR-Kit 4-pc cal kit, which uses a F-F adapter with dimensions, length = 32.5mm and includes Amphenol marking. Looks very much like:
Based on SDR-Kit calibration parameters, Short difference between Male and Female is 77.11pS one way.
Measured external surface to plane = 8.5mm (both ends)
Plane to plane = 32.5 - 8.5 - 8.5 = 15.5mm
15.5 * 3.3333 = 51.66pS
Vf = 51.66 / 77.11 = 0.670
Correction for Short cal device calculated Plane to short condition using 3.3333ps/mm is 11.4mm
Can't tell where the short condition plane is within the device, but 11.4mm between ref and short seems plausible.
On 5/22/20 5:23 PM, Kurt Poulsen wrote:
Hi John and David
Best 73 de Kurt de OZ7OU