Re: Nano VNA SAA-2N and G3XDY QRO filter
Just select 4GHz as the upper frequency limit when setting the sweep range and do a calibration run prior to measurement. Dynamic range and accuracy may be a bit below spec but it does provide a good indication of what is going on.toggle quoted messageShow quoted text
On 09/03/2021 11:23, Brian Howie GM4DIJ wrote:
In message <email@example.com>, Dave G1EHF via groups.io <firstname.lastname@example.org> writesThe VNA SAA-2 series use the ADF4350 as source oscillators, which inI got a SAA-2N recently . Is there anything special you have to do to get it to go up there ?
This email has been checked for viruses by Avast antivirus software.