Re: Agilent 346A and N9069A reading high?
Conrad Farlow <conrad@...>
Good points made by all and to mitigate against it I have used narrower bandwidths. I makes very little difference but it is still worth noting so thankyou for that. Basically the N9069A allows you to use any bandwidth that the EXA series of instruments can use, i.e very narrow indeed. I went down to 100kHz and yes I did calibrate at this bandwidth.
I think that there may be several issues.
I do have a screened enclosure big enough to do the
measurements in, it's not a chamber but I can get access to one
in Essex next time I visit G8CUB's splendid establishment.
I will investigate this further most probably after my much
needed holiday, the first for 4 years! I will be on an island in
the I8 region.
Thanks for the advice from all, it's nice to have the kit :)
John Makes a good point - NF meters can typically measure in bandwidths of several MHz and are assuming the noise figure, gain and match is constant across this bandwidth. When you design a preamp for VHF it is normally a good idea to make it narrow band, especially in the UK where we have very strong pager signals either side of 144MHz. To a DXer, there is nothing of importance outside the 144-145MHz range and a preamp may have a 3dB bandwidth of 2MHz or less to cover this range plus a bit. What this means depends on the filter shape and the NF meter bandwidth, but its quite possible to be missing nearly half the noise due to the filter. As John says this will mean the ENR calibration won't be right so the difference between noise on and off will be lower than expected resulting in a higher recorded noise figure. You can fix this with a filter which is difficult, or by doing the maths if you know the relative bandwidths, which may be easier. The negative noise figure issues is probably just noise. If you expect to get better than 0.1dB accuracy after calibration you will need to be very careful, do a lot of integrating and keep everything at a constant temperature and do it all in a screened room. You should see a mean value of zero. The fact it is reading -0.1dB consistently could just be the instrument needs recalibrating, or maybe you have not got the settings right for temperature, or indicate external noise is getting into it from local RF sources and disturbing the calibration. If you are doing this at home I would go with the QRM. Since they installed a mobile phone mast at RAL near the RF lab, we haven't been able to measure VHF/UHF noise figures accurately in the RF lab. Mike --- In ukmicrowaves@..., G3XDY wrote:Hi Conrad I have also seen abnormally high NF readings on an example of the HA8ET preamp. This preamp includes narrow band filtering, and it could be that the noise bandwidth of the preamp is narrower than the Agilent test set bandwidth. I recollect that you might be able to configure the Agilent box for a narrower measurement bandwidth, but I don't have the manual to check. The 8970A will suffer from the same problem and is not configurable, the only way to test such preamps with the 8970 series boxes is to build a low loss narrow band 144MHz filter, then calibrate the instrument with the filter in circuit and then test the preamp via the filter (bear in mind that the filter won't look like a flat 50 ohm source so there will be errors due to changes in the impedance seen by the preamp input unless you include an isolator after the filter). 73 John G3XDY On 18/07/2013 13:44, Conrad Farlow wrote:I have access to an Agilent 346A head with an N9069A noise figure measurement application running on an EXA series spectrum analyser. It seems to have number of issues. 1. After calibration the noise figure and gain are never truly zero, the noise figure normally reads -0.1 or similar and the gain will be -0.15 or thereabouts, these are just typical figures. 2.When I measure noise figure I see good results on the gain but the noise figure seems high. For instance my HA8ET Extra 2 LNA reads about 1.4dB whereas I am pretty sure that it is much better than this. I have Tcold set to the temperature on the surface of the device as measured by a thermocouple. I have used some loss compensation before the device, I have generously allowed 0.1 dB for a high quality female-female N type. Does anyone have any advice as to what is going wrong? I do have an 8970 that I could use for a reference check but it hasn't been switched on for a while, shocking I know....... Bit puzzled..------------------------------------ Yahoo! Groups Links <*> To visit your group on the web, go to: http://groups.yahoo.com/group/ukmicrowaves/ <*> Your email settings: Individual Email | Traditional <*> To change settings online go to: http://groups.yahoo.com/group/ukmicrowaves/join (Yahoo! ID required) <*> To change settings via email: ukmicrowaves-digest@... ukmicrowaves-fullfeatured@... <*> To unsubscribe from this group, send an email to: ukmicrowaves-unsubscribe@... <*> Your use of Yahoo! Groups is subject to: http://docs.yahoo.com/info/terms/