hope your well.
You coming to Martlesham??
I am not familiar with that meter but there are a couple of
1. the head has been overloaded (prob the reason)
2. the calibration of the unit is out.
I was looking for details of a Marconi 6559 meter on the net and
found a few places with handbooks. Maybe that will help.
Its likely that the meter has been overloaded and the head is now
misalligned - I have some details in the Microwave Updates on fixing
these thermistor type heads - I will see what I can find for you.
--- In ukmicrowaves@y..., "Howard Long" <hlong@b...> wrote:
I recently purchased a 2nd hand Marconi 6950 RF Power meter with aa
calibration job or is one of these bits likely to be dodgey, and