P6006 style probes...loose connection at the BNC?
Recently I got me a beautiful Type 547, cart, and all kinds of goodies at an estate sale. Naturally it makes a good home for the Type 1A4 plug in I already had, and it came with one P6006 probe. I got two more P6006 recently (need to find a fourth!) but one one of these new ones has an intermittent connection going on near the BNC, Is there any way to get under the strain relief to fix it there, or is it hosed?