Re: Tek 576 Curve Tracer Repair
Bob Koller <testtech@...>
toggle quoted messageShow quoted text
The machine left, or your left viewing from the front?
Sent: Wednesday, April 10, 2013 6:19 PM
Subject: [TekScopes] Tek 576 Curve Tracer Repair
I bought a 576 from eBay with a Tektronix cal from Dallas which expired 01-DEC-10. Everything was going well and I was learning how to use the instrument...until today.
I was attempting to characterize 6C33C tubes when smoke began to emanate from the top array of ventilation ports on the machine's left side. Immediately, I attenuated the Variable Collector Supply and switched the machine off.
I did not observe any POP or other sound in connection with this event.
Now that I've recovered, somewhat, my equilibriium -- I'm writing the post to ask this august group if there is anyone, any resource in my neck of the woods -- San Diego, that can take a look at my beloved Tek 576 CT.
Please, let me know.
Very best regards,