I finally tracked down a shorted 2.7uf tant on the timebase board and got that
replaced today. After putting everything together it booted up and passed all
the self tests. I have it hooked up to my SG503 right now and it appears to
be functional (yay).
Now it will probably take me a week or so to figure out how to actually use
it, assuming nothing else fails in the meantime.
Anyone else have one of these? I don't recall seeing any mention recently.
Paul Amaranth, GCIH | Rochester MI, USA
Aurora Group, Inc. | Security, Systems & Software
paul@... | Unix & Windows