Re: Resurrected 11402 and plug-in compatibility
Denis Cobley <denis.cobley@...>
Hitoggle quoted messageShow quoted text
The 7K units are vertical plug in 's only - no sampling, timebase or specan,
curve tracer plugs.
This is because the 11K slots are all vertical only - the timebase is
digital and will not accept any plug that produces a ramp for sweeping.
Also, you need to cut the rear plastic of the 7K unit (look at them side by
side to see the difference.
Then, when you have it in the scope you will need to select it manually as
the scope cannot "see" it.
I would just buy some more plugs on ebay.
----- Original Message -----
From: "madmanoneresurrected" <firstname.lastname@example.org>
Sent: Friday, August 20, 2004 10:04 AM
Subject: [TekScopes] Resurrected 11402 and plug-in compatibility
After grabing a 11401 off EBay and finding that it has 1 defective