I have a 465M scope and after a while the display went dead.
Examining the internals revealed the 1/4A fuse was blown and
continues to blow after replacement. I have a manual, but nothing
is said about causes for this fuse blowing. Without the fuse in
place, I get +55VDC from the power supply Cap side (+32VDC
unregulated), and +.5VDC from the HV transformer side (pin 4 of part
T550). Where there should be +95VDC, I get only 11.4 VDC.
1.What are the most likely causes for this fuse blowing, when all
was working fine.
2.What are the poosible causes for the +95Vdc to be way low.
Any responses welcome...