When measuring resistances with the DM40 I noticed that the two lowest
(200 & 2k) ranges would only display approximately half of the
resistance being measured. The rest of the ranges display correctly.
Pouring over the SM shows that the two lower ranges switch in a 1k
resistor but that resistor measures good. Anyone ever run into this
type of trouble and can maybe shed some light?
No trouble with any range on the voltmeter.
You can reply here or email me.
Tnx & 73