Re: Tektronix 2230 screen capture via plotter port? (was: Re: Re: DSA602 screencapture software...)
toggle quoted messageShow quoted text
I use a free program that will capture a screen dump via GPIB. I have been using this for years with my TDS744 and HP3577 network analyzer, and it does the job fine.
To run it, here is how I use it:
Set GPIB to “no user assigned address”
Press “w” (wait for device) or “W” (wait for device, then print the screen automatically).
On the scope, you just do a plot, and the software captures it.
Press any key to get out of the “wait” mode.
Press “p” to print, or “s” to save the screen shot.
I have no affiliation with the developer, I am just a fan!
From: TekScopes@... [mailto:TekScopes@...] On Behalf Of Brad Thompson
Sent: Monday, February 21, 2011 1:08 PM
Subject: Tektronix 2230 screen capture via plotter port? (was: Re: [TekScopes] Re: DSA602 screencapture software...)