Re: Tek 453 - Q913 Part Number
--- In TekScopes@yahoogroups.com, "nu7f" <jbstamper@c...> wrote:
I have an early (Serial 009869) Tek 453 scope with no trace. Q913before
I go digging into the HV side.----------
My microfiche RPR (replaceable parts record) says that 151-0133-00 is
replaced by 151-0220-00 at all occurrences in 453, all S/N's, except
for Q423, Q753, Q754, Q775 and Q785 where no replacement is listed.
You can find a scan of this page at