Re: p6032 Probe
Craig Sawyers <c.sawyers@...>
This is a cathode follower probe, intended for the 1S1 sampling plug-in. Ittoggle quoted messageShow quoted text
is good from DC to 850MHz, but goes beyond that. Depending on attenuator
head, it has a loading of 1.3pF (1000x) to 3.6pF (10x) in parallel with
10M. Load resistance at the GR connector is 50 ohms.
Look upon it as a big FET probe, and you will not be far wrong.
The power connector interfaces with a socket on the front of the 1S1, and
needs the following voltages:
A - ground
B, C - 12.6V at 180mA
D - +100V
These probes give a very clean response - I'll send you a picture of what
they are capable of.