Re: 2712 SA with display (and normalization) issues #photo-notice
Have you looked here?toggle quoted messageShow quoted text
Not OCR, but probably useful. Failing that, try Artek Manuals (http://artekmanuals.com/). They don't have it listed but they might be able to help.
From: TekScopes@groups.io [mailto:TekScopes@groups.io] On Behalf Of Ronan
Sent: 18 July 2021 22:59
Subject: Re: [TekScopes] 2712 SA with display (and normalization) issues #photo-notice
Just one more tidbit (before I decide if I'm going to dig deep into it).
Enabling debug diagnostics and re-running the Frequency Normalization indicates this:
“ERROR in MAIN DAC OFFSET: Cannot center start spur”
Explanation: "Beginning at 10 MHz/div span, the start spur was located and
centered, then span was reduced. In 1 MHz/div span, the signal
could not be centered within 30 horizontal storage bits."
I have a “oddity”, a +30dB rise of about 70MHz width:
https://ibb.co/db150Tr 1486 MHz
https://ibb.co/5BLWK3D 1498 MHz
Does anyone have an OCR of the 2712 Service Manual?