On Wed, Mar 3, 2021 at 07:00 AM, Ozan wrote:
Are you still seeing ~ -61mV as before at collector of Q1548 with no dot onI see ~42mV at collector of Q1548, no dot on screen in XY Mode.
Base of Q1544 is at ~1V, HFE measured with a TC1 tester : 214
CR1539 is ok : ~.6V and 0L
R1536 measured in circuit gives : 300kohm and 100kohm (when inverting the test leads of DMM).