Chuck Harris <cfharris@...>
Hi,toggle quoted messageShow quoted text
It is standard practice to reset the power cycle counter,
and the hours counter when a scope is calibrated.
The purpose of the counter is to give the calibration
technician some idea of how long it has been since the last
calibration, and how many power cycles.
You can read the counter by entering the diagnostics menu
and running EXER05. See page 6-13 of the service manual.
Ondrej Pavelka wrote: