Re: 2465 Ch1 attenuator repair question
Chuck Harris <cfharris@...>
The BNC is held in place by a set screw. It is easy totoggle quoted messageShow quoted text
fix. The biggest problem is taking it out, as you have to
remove the knobs from below the CRT, the bezel with all of
its screws, and unsolder a couple of parts from the hybrid.
Unfortunately, the set screw is on the side away from the
bottom of the scope, so removal is necessary.
If the connector is intermittent, it is possible that a wire
or solder joint has been broken from the motion.
saipan59 (Pete) wrote:
Hi, On my 2465, the Ch1 BNC is slightly "loose", and the signal is intermittent depending on which way the probe cable/connector tugs on it.