Note: groups.io will be down for maintenance on Wednesday, October 5th, starting at 9AM Pacific Time (4PM Wednesday October 5, 2022 UTC), for approximately one hour.
Hi, On my 2465, the Ch1 BNC is slightly "loose", and the signal is intermittent depending on which way the probe cable/connector tugs on it.
The connector seems to be integral to the A10 attenuator assembly.
Looking at the maint manual, I see and understand (I think) the instructions for removing the Attenuator. But I don't see any details on how the A10 is assembled physically.
SO, my question is: If I go to the trouble to remove A10, is it likely that I can make a repair to tighten up the connector and fix the intermittent connection? Or am I going to need a replacement module?
If appropriate, does anyone have an Attenuator that you can take a picture of (with the cover removed)?