Re: Tek 577 D1 Repair (B07xxxx)
4. (B108xxx,My 2nd 577 D1)
2020-Nov-2~8: Max 8 traces displayed on screen(9 lines included the baseline)
Checked the staircase signal:
Saw only 8 steps, 660mV /per step <-------NG (should be 10 steps,500mV/step)
Adjusted the step amplitude VR(A1 Board) and got the staircase of 10 steps (500mV/step).
5.(B108xxx,My 2nd 577 D1)
2020-Nov-9~15: The Collector supply disabled light turned on right at power on.
Vb=+1.3V <-------NG Vbe=+1.3-(-12)=13.3V>>0.7V
Fixed this issue by replacing the 2N2222 with a 2sc1384(50V,1A).