Re: 067-0525-01 or equivalent needed.
My guess (and it’s only that) is that they designed it to be enough better than any DUT that you wouldn’t ever worry about the fixturing. A good stress test, e.g, would be CMRR measurements at the upper end of the 485’s freq range. Such a measurement is sensitive to phase match.toggle quoted messageShow quoted text
Sent from my iThing, so please forgive brevity and typos
On Nov 11, 2020, at 13:27, Eric <firstname.lastname@example.org> wrote: