toggle quoted messageShow quoted text
I second the suggestion to use a J310. That's been my standard sub for general-purpose JFET buffers for a long time.
That said, I'm surprised that you didn't find joy with a 3819. The circuit is quite accommodating (necessarily) of a wide range of IDSS. The NTE133 should've worked, too; maybe it was just dead.
It's been some time since I took a close look at the 2215A's buffer, but a quick glance at the schematic says that the upper limit on acceptable IDSS is roughly 30mA. One that is around 10-15mA would be in the center of the zone, but the value is not particularly critical.
Prof. Thomas H. Lee
Allen Ctr., Rm. 205
350 Jane Stanford Way
Stanford, CA 94305-4070http://www-smirc.stanford.edu
On 11/8/2020 20:29, firstname.lastname@example.org wrote:
Got a Tektronix 2215A. Channel 1 was bad, no signal coming through. Channel 2 fine.
Found Q122 Jfet on channel 1 was a J210 part # and bad... already replaced a while back perhaps? should be orig J300 Jfet there?
After trying to sub Q122 without success with several Jfets on hand, including a NTE 133 that didnt work, decided to move Q222 (channel 2 same Jfet) to channel 1. This was found to be a J300 Jfet (never replaced?) and this made channel 1 work perfect and confirmed only prob on Channel 1 was indeed Q122.
Channel 2 is now without Q222. I tried to sub with several other Jfets I have on hand...k19...NTE132...Best result I had was a 2n3819 that works fine for about 1 minute then signal dissapears slowly. Freezing bring it all back. I suspect not a bad 2N3918, just freezing it make it behave more like a J300...that the circuit wants...
Can anyone confirm a few Jfet #s (still available? or not...I have large inventory of 70s-80s NOS parts...but no Jxxx Jfets) that will work reliably in that circuit? I think pretty much all that series of scopes used the similar input circuit. The Tek part # was 151-1124-00...
I have ordered some pn4392 Jfet but now thinking they will be no good in that circuit too...