Hi there! New to the group, I posted this on Reddit at /r/rfelectronics first and was referred to here.
I recently got a Tektronix 492 spectrum analyzer on eBay. I paid around $320 for it in a Best Offer, and apart from the issue in the title and the trace rotation needing some adjustment, everything I've been able to test on it does work, so I'm still pretty happy with the purchase.
That said, it does have one quirk I haven't been able to diagnose. It can operate as normal, but if left plugged in, eventually (after anywhere between ten minutes to a day or two), everything but the noise floor will disappear, as if the DUT was simply unplugged. This happens even with e.g. an antenna looking at broadcast signals, so the signal is certainly still there, it just stops seeing it. This doesn't require any kind of physical jolt (I don't think it's a connector being bumped out, for example), can happen when the instrument is on or off (you'll find out when you go to use it again), and can happen as the user is interacting with it or when it's just sitting there. When this happens, the device still "works" - you can look around the noise floor, change settings, all as expected, but no amount of disconnecting and reconnecting the DUT, making sure everything's plugged in right, or fiddling with the knobs will bring the signal back. The only fix is to either power cycle it with the front switch, off for 30 seconds, after which it will work for a few seconds, or to unplug the power cable for about 30 seconds and then plug it back in, after which it's back to normal for minutes to hours until it happens again and necessitates another brief unplug. This seems especially weird given that, as I understand it anyway, the power switch on the 492 is not a soft on/off - I'm surprised there's a difference between power cycling with the switch and power cycling at the plug. The failure is sudden - I've seen it happen, the input doesn't slowly fade into the noise floor, it disappears abruptly and sometimes in the middle of a trace. I haven't discounted that I may be doing something wrong - this is my first "real" RF instrument and I may be missing something obvious - but I'm not doing anything when it fails that doesn't work as normal any other time. I also don't think it's some out-of-spec input signal (too much power or DC on it or something) messing up the frontend, as the failure is temporary and I've seen it happen with just a normal, passive antenna connected. I don't hear relays click when it stops working, so I don't think it's accidentally enabling the attenuator or something.
Given the relatively easy workaround, this is more of an annoyance than a dealbreaker issue, but I'd still like to fix it. What could cause what's seemingly an input problem (given that everything but the actual measurement keeps going) but need a full power cycle and seemingly, time to discharge power supply caps, to resolve? Is there some always-on power rail powering whatever's failing that power cycling from the switch wouldn't affect? Is this a known problem?