There is some discussion in the FG504 manual that suggests
toggle quoted messageShow quoted text
that many TM50X frames were built with one of the power
transformer windings out of proper phase for the FG504 to
I would guess that the windings, being AC were just a pair
of unmarked wires, and the assemblers didn't know that they
needed to pay attention to which wires went to which pad.
It wouldn't be a problem most of the time, but in some of
the many possible configurations it could result in issues.
Harvey White wrote:
On 8/10/2020 4:13 PM, Stephen wrote:
I figured that the plugins that were not working before but are now were the onesActually, looking at the TM503A manual gives you an idea of the reason. The outputs
needing the NPN transistor.
The PS503 probably needs the PNP only.
of the supplies go through the frame transistors. However the drivers for those
transistors also provide a path for the regulated output. It's a very "weak" current
limited output, which would show up with a good meter, but the supply would not
provide anything near full current. The 5 volts uses a 7805 regulator, hence the 1
amp current limit.
The DC503A 5 volt ground is common to all modules. The common lead for the +/-
Now that all my plugins are powering on, I have to figure out 2 things. Not that
the 1st is a real issue, but I’m just trying to understand the concept.
1- While the unit is inside the 506, and when my DMM ground lead is connected to
ANY of the ground banana plugs can I measure the +/- 20VDC as well as the 5VDC.
But when the plugin is connected To the extender, I have to switch the ground lead
to the appropriate one on the PS to be able to read the voltages
supply is floating. I'd measure from the frame to frame of the power supply to see
what's going on there. However, a possible suggestion is to look on the backplane to
see if there any added wires.
I'd suggest backplane wiring, as in there shouldn't be any. If someone used this
2- I still don’t get why the DC503 shows only 1 digit or nothing at all when
plugged in directly into the 506, but is showing every single one, albeit some are
missing a segment, when used with the extender... A mystery to me. But I’ll
follow up on this in the appropriate thread I started on troubleshooting it.
with backplane wiring, then the slots become dedicated to a particular module. You
didn't connect the additional wires to bring out the plugin signals, where in the
frame, they're connected.