On Sun, Jun 14, 2020 at 2:18 PM zenith5106 <hahi@...> wrote:
On Sun, Jun 14, 2020 at 07:33 PM, Martin wrote:
Do they have the same F/W versions ?
But I wonder why this unit always executes a "new configuration" and "partial
enhanced accuracy" on startup. It have another 602 that is ready immediately
after self test
I seem to remember that being a F/W bug but at the moment I can't find any info
about which version it was. The latest version I know of, and probably the last,
is 2.23, same as the A-model.
Data point: my DSA602A with version 2.23 does this, my DSA602 with
version 1.3 does not.
Definitely a bummer when the final firmware version has a serious bug :(