How is a zener dissipating the power necessary to get the
toggle quoted messageShow quoted text
voltage down to 12V any different from a shunt resistor dissipating
the power necessary to get the voltage down to 12V?
Seems to me they are both the same, and both dissipating the same
power the nicad pack is taking in. Granted, a charging nicad pack
is endothermic, and a resistor is exothermic...
Tom Gardner wrote:
On 11/06/20 16:02, Mark Pilant wrote:
I tried for quite a while to get the "battery fooler" to work *reliably* inDitto.
the 1502 and 1503 I have. I was never successful.
The zener concept isn't too bad, but there is the power dissipation in the enclosed
I did try the Batteries Plus route and had them make up a replacement pack.I have found these acceptable
While it looked fine, it wouldn't fit. The original Tek design used flat
top "C" cells, and pretty much all that is available are button top "C"
cells. According to the specifications I could find, the "original" flat
top "C" cells were slightly shorter than the button top "C" cells; by the
height of the button.
The diameter is slightly too small, but that is easily rectified with some
BTW, I did some checking, and these batteries will last about a month beforeYes. There is an ~1mA (higher if the chart recorder is present) drain even with the
needing to be recharged when "sitting" in TDR that has been turned off.
front panel switch off. If the battery is left in then eventually one or more of the
cells will be damaged by being reverse charged. That's the reason the manual
specified the battery should be charged monthly.
Personally I just leave the battery pack out of the case. If I wanted to keep it in
the case, e.g. for shipping or long-term storage, then I would remove the fuse in the