Re: Powering P6201 or P6202A from Tekprobe
One other way is to use an 1101 or 1101A, perhaps an 1102 (not sure about that).toggle quoted messageShow quoted text
The 1101 is simply a box with 4 lemo chassis mount connectors on it, and a simple LM317 LM337 supply.
The 6201 and 6202A (IIRC on this) use only +/- 15 volts. The additional 5 volts on the extra pin of the 1101 is unused.
I think you've got a pretty decent scope collection, so if you have a parts donor 7000 series frame, it's going to have the female connectors on it already. Yet Another Project.....
To the other questions, I don't know the answer.
On 3/4/2020 4:58 AM, David C. Partridge wrote:
I own a P6201 and a P6202A. Neither will plug into a scope with a TekProbe