Re: Tektronix 2467B Auto Measure Faulty?
Chuck Harris <cfharris@...>
The Auto Measure mode only works with CH1 and CH2 inputs.toggle quoted messageShow quoted text
If both CH1 and CH2 are selected, CH1 takes priority.
In the Auto Measure mode, there is a button all the way to
the right called "CONFIGURE". In the CONFIGURE screen there
is a button all the way to the right called "PROBE"
In the probe configure menu you can select what happens when
the probe ring (identify) is grounded.
The first three options are IDENTIFY, AUTOCONFIGURE, INIT50%.
The last option is STEP. Step allows you to increment through
the preconfigured test setups for the scope..
The operator manual describes this very well, but if you wished,
you could program the test setups in the scope to measure FREQ.
I don't know what the default test setup is, but FREQ seems
Test setups are reached through the diagnostics menu.