Re: Tektronix 2467B Auto Measure Faulty?
You are looking at the military service manual... it is not the besttoggle quoted messageShow quoted text
manual to have. The US military standardized on the Fluke calibrators.
You can calibrate your scope with the usual SG503/504, PG506, TG501
and TD pulser arrangement.
Go to Didier's website, and get this manual:
leandro.lindemann via Groups.Io wrote:
The service manual states that a Fluke 5820A calibrator is required to calibrate the 2467. This instrument is inaccessible here in Brazil.