Re: 5116+5D10 intensity problem
You're welcome. I don't know if you have the particular HV transformer, but as Chuck (next) says, you're definitely looking at a low high voltage, so the question, of course, is why....toggle quoted messageShow quoted text
For the probe: That's likely a P6015 or the equivalent. It may be used with or without the dielectric liquid. WIth, I think that the rating is 30KV, without, about 18KV. You should be fine without it. You may want to try using it with an adaptor with a meter, but it is designed to work into a 1 meg input impedance, and most meters are 10. Paralleling the meter input to make it 1 meg is recommended if you try this. I'd also try a test run (if you do that) on a known 100 volt or so supply to make sure you get the required 0.1 volt on the meter. If not, then you're using it on a scope, and while you'll get a good idea, of voltage, it will only be within about 5% or so unless you have an A/D plugin. The 5D10 might just do for that if you could see the readouts.
I recommend hunting the internet for the manuals for everything you have, if you haven't already. I have an electronic library in several places. Some of that information is available for only a limited time.
On 2/17/2020 3:22 AM, Yeun-Jung Wu wrote: