Re: Anyone needing parts from Tektronix 310A, 531A or 533A scopes?
Brenda-toggle quoted messageShow quoted text
Consider saving the power transformers, as they have a tendency to develop a leakage path from the elevated potential CRT filament winding to whatever adjacent winding. Failed 310's are commonly saved by adding an external filament transformer, but that's a bit unsightly.
On Saturday, February 8, 2020, 4:59:49 AM EST, momemeca <momemeca@...> wrote:
Another request from the land of Oz :-)
I am restoring a number of older TEK scopes, including a 310A, and would dearly appreciate some parts if not a complete donor unit.
Perhaps the other gentleman (Morris I believe) could put me in touch with the fellow in AUS?
On 8 Feb 2020, at 10:57, Brenda via Groups.Io <email@example.com> wrote: