Re: Tek 533 -150 test point
http://w140.com/tekwiki/images/e/e1/070-0258-00.pdftoggle quoted messageShow quoted text
Page 62 shows the location, not sure if the terminals are marked or not.
OR you can use the body of the main filter capacitor for -150V (after
removing the protective sleeve)
You are correct, -150V is the reference for all, if you adjust it, rest all
will follow. Details in the manual above. Check calibration section.
On Wed, Jan 15, 2020 at 2:31 PM N7QL <n7ql.us@...> wrote:
Since I do not see this test point labeled can anyone tell me where it is?