First of all Harvey, Happy New Year 2020 and wishes of lots of health and
toggle quoted messageShow quoted text
great ELECTRONS in your life!
If it wasn't for you, my scope would not work, so I trust you and wish you
are my neighbor!
Anyway, thanks for helping everyone here, including me. ( tonysfun.com and
click on MY VIDEOS to see how micro-knowledge functions in my very small
brain! ... ha ha ha)
Lastly, I wish you all HAPPY NEW YEAR 2020 and POSITIVE ELECTRON'S in your
life and scopes!
On Sun, Jan 5, 2020 at 10:35 PM Harvey White <madyn@...> wrote:
There are some tests you can do with a partially working scope, but you
are going to be very limited.
1) assuming horizontal sweep problems, you use an external ramp or the
"other" sweep if you have a dual sweep model.
2) if you have a horizontal amplifier problem, using the scope in XY
mode, even with the calibrator, can give you some idea of whether or not
it's the sweep or the amplifier.
3) if you have a dual channel scope, you can use a working channel to
debug the separate part of the vertical amplifier
These generally end up being go/no-go tests and are somewhat crude.
Function generators can make a very crude horizontal sweep. If they're
gated, then a gated ramp might work as a sweep.
Very crude, but better than nothing.
Generally, unless you have a scope specifically designed for self
diagnosis, you're better off with another, completely working, scope.
Even then, a scope designed for self diagnosis makes certain assumptions
about what's working, especially once you get past the "is the CPU
On 1/4/2020 4:29 PM, Panos wrote:
I have read enough service manuals for the oscilloscopes, and till todayI didn't find any of them to mention, that someone can use the same faulty
scope (partially or totally) to repair his self.
But as always we can hope for something like that. After all, don'tthey say that hope dies always last? :-)
you want really to repair it, then proceed it with another oscilloscope.
In my opinion if you have time to play then do it by that way. But if