Re: 2445 CH 3 and CH 4 vertical setting - volts/ div where ?
You only have two choices, 0.5 V per division and 0.1 V per division by the two small button / switches beside the position controls. Of course, with a 10X probe, you can multiply that. Besides good trigger channels, they are good logic channels.toggle quoted messageShow quoted text
----- Original Message -----
From: "Tom Gardner" <tggzzz@...>
Sent: Sunday, November 10, 2019 4:08 AM
Subject: Re: [TekScopes] 2445 CH 3 and CH 4 vertical setting - volts/ div where ?
On 10/11/19 02:36, vaclav_sal via Groups.Io wrote:Just adjured 2445 scope and trying to get familiar with the controls.Most Good Things are available on the w140 TekWiki, in this case