Re: 2465A channel 2 problems
Hi Gerry,toggle quoted messageShow quoted text
First, you are not supposed to get a trace, just a pair
of dots, approximately 10 divisions apart.
This is a crucial step, as it affects all future steps.
I would suggest that you always put it in VAR mode, as it
will always give you an adjustable range to hit the mark.
But note: It is most important that the VAR knob be back
in its click before you start the next, and all subsequent,
steps. If you do not do this, you will achieve out-of-range.
If you repeat this step after having done it recently, you
will usually find that the dots are perfectly spaced.
There is no linkage between the 50 ohm annunciator light and
the 50 ohm relay in the preamp. Unless your preamp is broken,
it is in 50 ohm mode.
As a test, when the scope is in normal mode, put the preamps
in 50 ohm mode, and measure them with an ohmmeter. If they
are other than 50 ohm, you have the ever popular burned out
50 ohm load. Unless you are very clever you will be replacing
your preamp, as the resistor is printed on the ceramic substrate.
Another question or two for those in the know: At the beginning of the Cal 02 procedure, it states to connect a 0.5 V standard-amplitude signal to CH 1 and to use CH 2 Position to vertically position the "trace" to within 1 division of the center graticule line, and then to use the CH 1 Position and Volts/DIV VAR controls to obtain a 10-division horizontal signal. I don't get a "trace," only 2 dots. I can position the dots with the controls to put one dot on the first graticule line and the other on the 10th.