i have a reak tek 1a7a manual that came when i purchased my 1a7a maybe i could scan a little of it for you
toggle quoted messageShow quoted text
----- Original Message -----
From: "bfoeken" <email@example.com>
To: "arthurok" <firstname.lastname@example.org>
Sent: Friday, December 02, 2005 12:03 AM
Subject: Re: [TekScopes] Tektronix 1A7 and 1A7A Plug-ins
Thank you for the information.
I have a faulty 1A7A that has problems. I haven't spent alot of time fault finding it yet and I only have the 1A7 manual (BAMA) as a reference.
I assumed the general layout would be similar but circuit references could be different.
All I have determined thus far is that pin 3 of connector P11 (vertical output signal) has 65.2VDC instead of 67VDC volts as indicated in the 1A7 manual.
Pin 1 of connector P11 has the correct voltage (67.5V DC)
Therefore there is an approximate 2.5VDC signal feeding the Vertical amplifier in my 535A Scope. I am not sure what effect this will have as I have very limited experience in fixing oscilloscopes but I believe this is sending the trace on my CRT off screen (up).
I cannot get the trace to appear on the CRT even adjusting the vertical position pot through its full travel.
I think I will need to find a copy of the 1A7A manual.
Thanks again for you help
---- arthurok <email@example.com> wrote: