Re: Tek 475.
James R. Bartlett
Hi Michael.toggle quoted messageShow quoted text
Scope is working again.
Changed C963, replaced R963,.
Had to readjust the pot,for the + 50 voltsupply, as +15 was showing +18,
Trace seems reasonable and all controls appear to work.
Probably needs a good going over but the main thing is it is working.
Many thanks to yourself and others who came up with various suggestions.
Hope you have a nice weekend and enjoy Labor Day.
Again thanks for all
In the land of the Leprechauns 😁
On Fri, 30 Aug 2019 at 20:20, Mlynch001 <@mlynch001> wrote: