Re: Replacing Power LED/light on TEK 2465BCT
For what it is worth, the little green indicatortoggle quoted messageShow quoted text
is mechanically very fidgety and delicate, and
is often broken on the scopes I get for repair
The problem is a very tiny push arm moves the
shutters that hide the green spot, it breaks,
and is too small to glue.
The only way to reliably fix it is to replace it.
Ken Winterling wrote:
I have a 2465DVS which is essentially the same scope. It, too, has the mechanical green indicator built into the power button.Ken WA2LBI LG G6 ------ Original message------From: Tony FlemingDate: Sat, May 25, 2019 10:42To: TekScopes@groups.io;Cc: Subject:Re: [TekScopes] Replacing Power LED/light on TEK 2465BCTMine 2465DMS has no light - it is a switch - when you press it in, a green