Majdi S. Abbas
On Fri, Mar 01, 2019 at 06:11:50PM -0000, David C. Partridge wrote:
Specifed MeasuredThe -15V worries me a bit -- I've seen a shorted tantalum
around 40 ohms on this rail.
The character generator board has 1 uF tantalums directly
across the -15V and -50V rails; try disconnecting it and see
what happens with your -15V measurement.