Re: 2445A calibration
Hi Maxim,toggle quoted messageShow quoted text
The normalizer is just a resistor and a trimmer capacitor
in parallel, but in series with the center conductors of
a male and a female BNC connector. It is built in a little
You can do the same thing with a scope probe, using the
probe's trimmer capacitor to square up the waveform on
one channel, and the C205 to square it up on the other.
I doubt it needs adjustment, assuming that you haven't moved
That CAL03 throws a LIMIT on CH3 and 4 makes me even more
certain that your problem is your generator.