Re: 2465B SN 55xxx with TEST 05 FAIL 40
Tektronix wrote high option numbers, like OPT46, to show things,toggle quoted messageShow quoted text
like including 4 probes, when they usually only included 2, or
including a carry pouch, or adding a Japanese mains power cord,
..., stuff like that.
Other options were mnemonic to what the option provided, like
OPT14 being for P14 phosphor on some scopes.
However, I don't see 46 listed in any of the literature I have.
Right you are, only 4 SMD problem caps total. I need to have the board in front of me for facts to start falling in place. Definitively will carefully chop apart the caps before attempting removal. I did not intend to disturb any tantalums, unless really necessary. I've also read that the 20K trimmer pot near the DAC sometimes gets also damaged and needs to be replaced before the ref voltages are correct again. Maybe just order it too along the caps, just in case.