Re: Tek 7603 Followed Me Home
My guess would be the -50V unregulated supply'stoggle quoted messageShow quoted text
filter capacitor is no longer any good.
His VAC readings are certainly Vrms as read by his
DVM, so they are 2 x 1.4Vp-p, or in other words, he
has: 13Vrms x 2.8Vp-p/Vrms, or 36Vp-p ripple on a 50V
unregulated supply. No way that can work!
John Griessen wrote:
On 9/9/18 10:11 AM, Robert Hay wrote:I'll re-check the voltages50 and -50 being actual 44 and -44 is the big clue.