Re: Focus on 475 scope
The mesh dome is generic, one is as good as the next.toggle quoted messageShow quoted text
It does not wear out.
The 475 does not have a nice sharp beam. It is inherently
about 1.5 minor divisions wide.
It is at its absolute best when the bandwidth limiter is
turned on to its lowest bandwidth setting.
If the glass screen, or the polycarbonate blast shield,
is dirty, the focus can appear even less sharp than
The 475A is a little better, as I recall.
Is this mesh dome inside ? Maybe i could swap the Crt's and see if problem follow the crt.