On Sat, Dec 30, 2017 at 09:45 pm, <firstname.lastname@example.org> wrote:
What are the risks? Any calibration impact?Documentation is included with the zip file. Specifically, Page 18 of the Field Upgrade Procedures:
"When signal-path compensation has completed, the oscilloscope is in the same state of calibration as before the loading of the new firmware."
Sounds like you should be relatively safe!
Still can't find my GPIB board, so I'll have to wait to try this... Anyone have a NI GPIB/TNT ISA board, they wanna trade something for (or possibly sell dirt cheap)?