Stan or Patricia Griffiths <w7ni@...>
Stan (or others), you have helped me with this scope just recently. IHi Terry,
Here is an educated guess. I suspect you have two problems. The drift is
probably due to a bad nuvistor in the Ch 1. You might just try reseating it
in the socket.
The lack of detent in the Ch 2 Variable Volts/Division control could be due
to a broken shaft coupling. I can't remember if there are shaft couplings
here or not. This may be a real tough one to repair because it will be
buried under the CRT shield. I might have a part for this if you can
determine what you need and can get to it. If this turns out to be one of
those white plastic pots with a crack in the part with the set screw in it,
one way to fix it is to back the set screw out and epoxy the plastic. You
have to let it sit for a couple of days to make sure the epoxy is real hard,
then retighten the set screw.