I have no idea what the original diodes were other than fast recovery
toggle quoted messageShow quoted text
From the looks of the design, the MUR160G should work fine as a
replacement for all of them. It will not run any cooler however.
On Mon, 14 Mar 2016 22:18:53 +0000 (UTC), you wrote:
Those diode belong to Hitachi V-1060 label as D1013,D1503,D1504,D1531,D1541. All located at high voltage area.
ALL mounted 2cm above PCB surface but still show overheating sign. This unit is Fan less and very basic without readout.
Recently gwt another 2247A with much advance features. Plan to recap both and intent to upgrade those diode make it run cooler
On Tue, 15 Mar, 2016 at 5:49, David email@example.com [TekScopes]<TekScopes@yahoogroups.com> wrote:
What oscilloscope model and exactly which diode is this?
On Mon, 14 Mar 2016 15:18:39 +0000 (UTC), you wrote:
Hope you can advice me on this for my another scope using diode AU01,AU01,RU3AMLF-B1 Base on service manual.
Could all those replace with MUR160G as well?
Because it look char and going to burn anytime