Leanna Erickson <lle@...>
But now Murphy came calling as I lost all triggering, so have to back up
From: TekScopes@... [mailto:TekScopes@...]
Sent: Friday, January 09, 2015 3:09 PM
Subject: RE: [TekScopes] Video update 1/9. Tektronix 475A x-y discussion and
On the collector of Q904, I have -0.865 volts when I should have +0.5volts.
In the 465 and 485 scopes the A and B sweeps are identical. If the 475A is
the same, you can try switching components from one to the other, one by by
one, and see which one causes the problem.