toggle quoted messageShow quoted text
Thanks for the response.
I'd really like to understand what's causing this issue & what to do about it.
Presently I have the Trimble GPS 1PPS 2.8V logic 1mS pulse driving a level shifter to a 5V logic level and also providing the needed inversion for driving a non-retriggerable 555 one shot. The 555 is giving a 112mS long positive going 5volt pulse for every 1PPS from the Trimble GPS. I'm applying the 112mS pulse to the GPS input on the ProgRock. In this configuration, register 2 is being updated to a value that causes the output frequency to run low of the desired by ~18 Hz.
If I remove the 1PPS, ground the 1PPS input on the ProgRock & reboot the ProgRock, then manually dial in register 2 to give the proper output frequency, it'll stay there within a couple of Hz. over a period of many hours if not days, thermal effects not withstanding.
It seems to me that under GPS discipline the frequency of the crystal is being mis-read, or there's some magic in register 28 that needs to be adjusted; these are my guesses at this point.
My primary reason for the 555 is to allow me the opportunity to experiment with different pulse widths. Since the QLG1 gives 100ms, I started with a value near that. I can trim it to be spot on later, but 112mS is where my particular RC value came to land late last night. I know from experience that 1mS doesn't work, nor does 11mS, at least not as my ProgRock sits currently.
My QLG1 and the OCXO from QRPLabs are in the mail. Once I have those we can see if anything changes. If you have any further suggestions, I'm all ears.
Robin Midgett K4IDC
Hi Robin, Graham, Sid, all
There have been a lot of posts on this topic and being on vacation at the moment I kind of lost track of where this got to.
ProgRock should work as advertised on the web page and documentation. If it doesn't, something is wrong.
A GPS disciplined ProgRock should be accurate well within 0.05ppm (note that this parts-per-million specification means that the absolute precision in Hz, scales with frequency). This is what I have seen in all my tests and has been verified by other constructors too. To get the system working does require:
1. GPS correction threshold register must he understood properly. To get maximum precision requires setting the GPS correction threshold register to zero (default is 5Hz).
2. The 1pps signal needs to be at 5V logic level. Many GPS units have 2.8V logic level outputs and need level conversion. This can be done with pull-up resistors with some care and some inevitable degradation in noise immunity. Or with proper level conversion.
3. The GPS pulse needs to be long enough for the processor to handle it properly. I'm not actually sure what the minimum is. Many common modern GPS modules have 0.1 second positive pulses, these work fine. Some modules with very short pulses e.g. 1 microsecond wouldn't work.
Note that the QRP Labs QLG1 GPS kit http://qrp-labs.com/qlg1
was used in all my testing and works very well. It has 0.1 second positive pulse and it has proper logic level conversion. The QLG1 has a very accurate 1pps specified with max 11 nanosecond r.m.s error.
If the ProgRock isn't performing as expected then something is wrong. That could be an electrical problem (soldering problem, dead component etc); or some issue with the characteristics of the 1 pulse-per-second signal input.
If there's any question I haven't covered here, please let me know.
73 Hans G0UPL
On Mon, Jul 15, 2019, 16:31 Robin Midgett <K4IDC@...
Hi Graham & the group,
The issue Sid & I and possibly others who aren't reading the mail on this subject isn't about stability with the TCXO or OCXO. The stability with those optional oscillators is well documented and not in question, and for a broad range of applications, mine included, well more than sufficient.
The issue here is making the ProgRock work reliably with GPS discipline, and why that isn't happening reliably.
Diatribes about the NEED for GPS discipline in anyone's application are irrelevant simply because the fact is the product doesn't perform reliably as advertised with GPS discipline, and, very importantly, within the context of kit building and the price of the kit, this is not a deal breaker or terribly surprising. This is part of the value of kit building; the builder has the opportunity to improve the kit as they see fit, or not.
What is perhaps more interesting and worth studying is why the GPS discipline isn't as reliable as it should be, and what's to be done to mitigate that deficiency. Based on the current responses to our (Sid & my) posts, Hans may be the only person qualified to really answer the questions regarding the apparent deficiency.
Further, personal attacks are not helpful,
Andy Brilleaux. Let's leave those for the elementary school playground. Let's approach this issue scientifically and see if a successful solution can be had. I suspect it can and will.
Robin Midgett K4IDC
I have built a few prog rocks. I did not use the 27 mhz crystal as supplied, but installed a TCXO as suggested in the manual. I find ,even without GPS control that it is very stable. The pads are already there for the TCXO.