FWIW I have my com port set to 115200 and ACLog CAT works fine.
N1MM's Spectrum display works well along with CAT on the same port
(in cw). This is with an IC-7300.
toggle quoted messageShow quoted text
73, George K4GM
On 4/15/2020 4:56 PM, Scott Davis via
Thanks for your e-mail. I'm sorry that you have run into
trouble. I've had lots of good reports on the 7300, so I am
sure the software is fine. You will find the detailed
settings other 7300 users have sent along here (there is even
a video specific to the 7300):
Note that no other users reported the baud rate you are
Serving the Amateur Radio community with contesting
and general logging software since 1997.
From: Scott <scolit@...>
To: N3FJPSoftwareUsers <N3FJPSoftwareUsers@groups.io>
Sent: Wed, Apr 15, 2020 4:05 pm
Subject: [N3FJPSoftwareUsers] Rig Control Issue with ICOM
I want to use ACLog while working
POTA. My COM10 is set at 115200, so I can use the N1MM
Spectrum when contesting and it works well. I also use a
program called VSPE and the virtual port is COM20 (not too
important). I have the Baud Rate set with the "Other" at
115200. The Parity is None, Data bits 8 and Stop Bits 1.
Connection power set to RTS like in N1MM. Mode determined
by Rig and the Command to Read Frequencies is FE FE 94 E0
03 FD and the Command to Read Mode is FE FE 94 E0 04 FD.
The rig keys and Polling Rig, however it does not read
back the Frequency/Mode. If I click Frequency and send, it
will QSY to that frequency. So I'm close, can someone give
me some hint what I'm doing wrong?