Re: TP-1 SNR.
toggle quoted messageShow quoted text
Thanks for that.
Actually you suggested this during a 'bad signal' patch a year or so ago.
I did look at it, the 'building' and configuration of the programme and felt it was beyond my ability and didn't want endlessly to have to ask 'how do I set this up.'
The other thing would to only have a scale detailing 12.5 dB to 14.4 dB.
Today it's cool and cloudy here with the dish and LNB 'cold' and SNR is down to usual morning/day lower level 13.6 dB - quite steady at the moment, 09:43 UTC.
Power is -26 dBm which is the cooler evening reading.
In a message dated 09/07/2018 19:45:12 GMT Standard Time, gm8arv@... writes: